The performance of the PrenaTest® Option 1 has been validated in a clinical study. The study results of the maternal plasma samples (n=966) demonstrated a positive percentage agreement (PPA; equates to sensitivity) of 100 % (lower 1-sided 95% confidence interval of 91.8 %; n=35/35) and a negative percentage agreement (NPA; equates to specificity; n=931/931) of 100% compared to NGS-based PrenaTest®. The negative predictive value (NPV) was 100 % (lower 1-sided 95 % confidence interval of 99.68 %). The average fetal fraction of all examined blood samples was 8.1%. In 54 blood samples with a fetal fraction below 4% and as low as 2.4% a correct test result was achieved. Based on the 100% positive as well as negative percentage agreement of the results of the validated qNIPT concept (PrenaTest® Option 1) compared to the PrenaTest® based on next generation sequencing (NGS), the test quality for the PrenaTest® remains unchanged. The PrenaTest® Option 1 can be applied in the case of singleton pregnancies.
Stumm et al. 2012. Noninvasive prenatal detection of chromosomal aneuploidies using different next generation sequencing strategies and algorithms. Prenatal Diagnosis 2012, 32, 569–577
Stumm et al. 2013. Diagnostic accuracy of random massively parallel sequencing for non-invasive prenatal detection of common autosomal aneuploidies: a collaborative study in Europe.Prenatal Diagnosis 2013, 33,1 –7
Groemminger et al. 2014. Fetal Aneuploidy Detection by Cell-Free DNA Sequencing for Multiple Pregnancies and Quality Issues with Vanishing Twins. J. Clin. Med. 2014, 3, 679-692; doi:10.3390/jcm303067
Floeck et al. 2017. Non-invasive prenatal testing (NIPT): Europe’s first multicenter post-market clinical follow-up study validating the quality in clinical routine. Arch Gynecol Obstet DOI 10.1007/s00404-017-4517-3
Cirigliano V, et al. 2017. Performance of the neoBona test: a new paired-end massively parallel shotgun sequencing approach for cell-free DNA-based aneuploidy screening. Ultrasound Obstet Gynecol. 2017 Apr;49(4):460-464. doi: 10.1002/uog.17386. Epub 2017 Feb 28.
Illumina VeriSeq NIPT Solution Package Insert, Document# 1000000001856 v04, April 2018
A total of 469 samples were tested, of which 175 (37.3%) met the quality criteria. Three positive samples with a 22q11.2 microdeletion were correctly determined (3/3, 100%). All negative samples were correctly classified (172/172, 100%). There were no false-positive or false-negative results.
In a final internal blinded study, 20 samples from Phase 1 were examined. All samples were classified correctly. Due to the low number of cases a concrete test sensitivity and specificity cannot be derived.
Hofmann et al. 2014. NIPT: Welche Unterschiede zwischen den Tests gibt es tatsächlich? FRAUENARZT 55 (2014) Nr. 11
Groemminger et al. 2015. The influence of low molecular weight heparin medication on plasma DNA in pregnant women. Prenatal Diagnosis 2015, 35, 1–3
Wolf et al. 2016. Puriﬁcation of Circulating Cell-Free DNA from Plasma and Urine Using the Automated Large-Volume Extraction on the QIAsymphony® SP Instrument. © Springer International Publishing Switzerland 2016 P.B. Gahan et al. (eds.), Circulating Nucleic Acids in Serum and Plasma – CNAPS IX, Advances in Experimental Medicine and Biology 924, DOI 10.1007/978-3-319-42044-8_33